Orthoscan FD Pulse No output
January 28, 2020 12:36
Have a carm no image no output, when you first fluoro the meter senses radiation then goes flat black image and ramps to max on screen but no output. Have tried another tube, umbilical cable, and reloaded software and calibrations. Talked with Orthoscan they can't think of what may be causing it. Any ideas where else to look at on the unit without buying a bunch of parts.
to rate and post a comment
re: Orthoscan FD Pulse No output
February 07, 2020 10:24
I am the technical founder of Orthoscan. You can find our info at https://www.minicarm.com/ send me an email or give me a call.
INTUITIVE THINKING • INTUITIVE WORKS • INTUITIVE SOLUTIONS • Contact TIBS today for your GE OEC C-arm Sales, Service or Parts. Parts & systems in stock now. Exchange or sell us your old C-arm. Call 866.499.3966 or email email@example.com.