Hitachi Aperto Lucent Signal Underflow
July 24, 2017 10:38
Hello. The problem described bellow appeared several times before, but now it became constant.
During prescan next mistake appeared: 22034 Signal underflow (Coarse frequency search). Please check the setting of patient, Connection or the Mode of receiver-coil.
And after several attempts to make scan appeared: 22032 Resonance frequency can not be detected (Coarse frequency search). If the system is still in error then please contact the service office.
It appeared on any receivng coil.
So now on any tests (like central frequency search etc.) we get one of these errors. One time we got 220030: Signal underflow (TxGain). Please check the setting of patient and receiver-coil.
But most of time first and second errors occur.
Did anyone encountered such issue?
to rate and post a comment
re: Hitachi Aperto Lucent Signal Underflow
July 26, 2017 11:55
contact me at email@example.com. i can help you.
September 14, 2018 09:09
did you solve the problem?
Please send me your email. I need help
re: re: Hitachi Aperto Lucent Signal Underflow
February 03, 2020 10:07
Hii, did you face signal flow warning? What was the problem from?
September 14, 2018 09:08
ElsMed engineers are located in four different countries. 24 hour response time to any service call. Service agreement provided for end-users range from a 6 month period to a 3 year period and covers all system spare parts.