CT somatom definiation
December 23, 2019 10:20
hy all, i have a problem with somatom definition dual source ct. seems like tube A tests are failed and on TUBE B system is exposing. as per software configuration we can select one tube only if tube A is ok. approximately 6 month ago same error was poped up but siemens guy came over and he configure the system on single source i dont know how then software hone crash and installing new software the error came back can any one help me out on this .
1 how can i configure it on single source when TUBE A is not exposing.
how can i tell system or swap with A or B because my tube B is working v fine. code is CT _gsa_79.
also an i get documents
to rate and post a comment
re: CT somatom definiation
March 30, 2020 10:27
It's not possible to config single tube mode when Tube A defective. But you can exchange A with B physically.
Altima will inspire the trust, loyalty and commitment of our customers through positive relationships and collaboration, top quality equipment services, and a clearly superior value proposition. Call Today (469) 644-7063