February 03, 2020 09:56
BV Endura Error M032 Can you help me
to rate and post a comment
Doscar Ernesto Perez Pereyra
re: BV Endura
February 04, 2020 10:27
Error M032 are parameters reset. Need Bv Scope for take out the error with the calibrations.
February 04, 2020 10:29
032. Adjustment status: not verified.
Logfile text: X-ray tube performance failed or never performed. Perform the following actions via BV Scope: Install
- adjustment - X-ray tube - Filament. Install - Performance Test - X-ray tube performance.
re: re: BV Endura
February 05, 2020 10:18
Yes this is correct, but mostly when you have this error then the system has more adjustment problem.
Then the Standcontroller will not only have no Tube data lost.
February 04, 2020 10:30
You need to complete a successful performance test.
Our engineers are trained strictly on Siemens & GE PET to Guarantee an expert on all service calls. Our large inventory of PET & PET/CT systems & parts
allows us to provide a TRUE Turn-key solution. Ph:866-645-7457