Here is a Nikon Eclipse TE200 Fluorescence Phase Contrast Microscope
Setup for Brightfield with Phase Contrast and Fluorescence
This Nikon Eclipse TE200 Fluorescence Microscope is in good working order with all aspects tested and working fully.
Microscope optics have been cleaned professionally by our technician under a class 10,000 clean room hood.
Please see the exact video and photos below.
Testing we performed:
The Nikon Eclipse TE200 Fluorescence Microscope passed all tests and was found to be fully functional:
Tested halogen lamp & source fully: PASSED
Tested mercury lamp & source fully: PASSED
Tested port splitter: PASSED
Tested mechanical XY stage to limits: PASSED
Tested objectives for clarity: PASSED
Tested eyepieces for clarity: PASSED
Tested phase contrast condenser/prisms: PASSED
Tested fluorescence light path & shutters: PASSED
Tested filter cubes and turret: PASSED
Tested mechanical nosepiece: PASSED
Tested live feed quality: PASSED
Tested connection to computer: PASSED
Tested all power switches and light adjustments: PASSED
During testing and cleaning this microscope was found to have no issues.
A clear image was produced through the eyepieces as well as through the camera port.
All of the mechanical components moved freely and the test sample was able to be focused without excessive effort.
System consists of:
Nikon Eclipse TE200 Inverted Microscope
Binocular tube head
(2) CFI 10x/22 Eyepieces
ELWD 0.3 Condenser
Prisms: PhL, Ph1, Ph2, Ph3
Brightfield Light Path:
TE-DH100W Illumination Tower
Fluorescence Light Path:
TE-FM light path w/ manual shutters
Fluorescence Filter Cubes:
Nikon TE-FMC 4-position sliding turret
41018 GFP- EX: 470/40nm DM: 495lp EM: 500lp
TE-DH100W Illumination Tower
LHS-H100P-1 Lamp House with built-in power supply
LH-M100CB-1 Mercury lamp with Chiu Technical M100T Mercury power supply
10x/0.30 Plan Flour PH1 DLL
20x/0.75 CFI Plan Apo VC (this is a high quality Nikon APO CFI60 Objective, it was originally installed in a custom piece of equipment where manufacturer did not want to show that they used Nikon objectives)
40x/0.75 Plan Flour DIC M
AmScope MU300 3.1MP digital color camera
Computer and Software:
Dell Latitude i5 laptop running Windows 7 Pro with AmScope software
Communication and power cables.
Configuration and Setup:
Illumination Type: Brightfield, Phase Contrast & Fluorescence
Objectives: 10x/0.30 Plan Flour PH1 DLL, 20x/0.75 Plan APO VC, 40x/0.75 Plan Flour DIC M
Light Sources: 6V 30W Halogen Lamp, 100W Mercury Lamp
Stage: Mechanical XY Stage
This Nikon Eclipse TE200 Fluorescence Microscope is currently set up for Brightfield viewing w/ Fluorescence and Phase Contrast studies.
The MU300 AmScope CMOS digital color camera is capable of taking detailed digital images.
The system includes a Dell Latitude i5 Laptop to control the AmScope software for live images.
In this video you see the Nikon Eclipse TE200 Fluorescence Microscope fully assembled.
We then show you the power supply boxes and then show the TE200 microscope and its connection with the laptop.
You then see the sample being put into focus and then you see all the different components and connections of the microscope itself.
TO VIEW ADDITIONAL PHOTOS, VIDEOS AND VERY DETAILED INFORMATION SUCH AS TEST STEPS PERFORMED, CONFIGURATION, ETC., PLEASE VISIT OUR WEB SITE. TO GET THERE PLEASE CLICK ON "WEBSTORE" TO YOUR RIGHT AND YOU WILL SEE LINK TO OUR SITE IN THE MIDDLE OF THE SCREEN. THEN USE LAST 5 DIGITS OF MPN NUMBER IN THE SEARCH FIELD TO LOCATE YOUR ITEM.
DO NOT HESITATE TO CALL US OR EMAIL US WITH ANY QUESTIONS YOU MIGHT HAVE.
ALL OF OUR ITEMS ARE AVAILABLE FOR INSPECTION AT OUR FACILITY.
Items are sold as-is with no returns or refunds available unless explicitly stated.